RET system applying the high frequency of deep heat wave technology, the heat from inside to outside, and there is heat energy output both RET handpiece and electrode plate. To make the heat in the body part of heating time can be longer. Deep heat inside body can improve the ability of the body metabolism and reduce fat more effective.
Radio Frequency Slimming Treatment JF278 structure
1.Clean treatment parts.
2.Choose appropriate work head according to treated parts.
3.Apply gel on the electrode plate and make sure the gel covers the surface evenly.
4.Set work time.
6.Smear gel or creams on treatment part.
8.Clean face after operating.
9.Turn off the general power switch and unplug the power supply line.
10.Sterilize electrode probe with alcohol, clean electrode plate by clean towel.
1.New RET technology: a special high frequency for fat cellulite burning.
4.Not harm skin.
6.Aim at fat location to avoid heating other area.
1.The treatment parts should be smeared gel or creams, which is helpful for the electric wave heat energy to transmit, and it has better result.
2.The output intensity should be regulated from low gradually to high energy according to endurance of customers.
3.During operating, the probe of handle should be clinging to the skin and move smoothly, do not stay on any part without moving.
4.The system will pause automatically if the setting time is over.
5.Clean the leftovers on skin with warm water.
6.Please do not operate the machine on the electriferous floor.
7.Other people do not touch the electrode plate when the machine is operating.
8.Other people do not touch the operator and customer during treatment time.
9.The operator and customer do not be with the metal decoration during in treatment time.
10.The electrode probe is moved on the skin by the circulation way.
11.Prohibit use of the equipment in the following environment
---Hot and humid environment.
---Environmental with much smoke or steam.
12.Do not make the machine close to the wall, keeping a 30cm space around for cooling.
13.Not suitable for common fat; not use for face, chest, muscle and bony areas.
1.People with heart disease or high blood pressure, or configured cardiac pacemaker.
2.Patients with acute inflammation, asthma, deep vein thrombosis, thyroncus, cancer.
3.People with hemorrhagic disease, trauma or who is bleeding.
5.Medical plastic parts, or parts with artificial in filling inside.
6.People with metal inside the body(include metallic tooth).
7.Patient with skin inflammation or with edema.
8.People with an abnormal immune system.
9.Numb or insensitive to heat.
10.Drunk or high fever patients.
11.Patients with sever infection.
12.Children, please use under the guidance of professional.
Please make sure your contact information is correct. Your message will be sent directly to the recipient(s) and will not be publicly displayed. We will never distribute or sell your personal information to third parties without your express permission.